The NexTemp Clinical Thermometer is individually wrapped and intended for single patient use. It is designed for high accuracy readings and to prevent patients from exposure to infections such as C. difficile, MRSA, Norovirus, and Rotavirus. NexTemp thermometers circumvent this risk by providing a clean, sterile instrument that is used with a single patient and then discarded. Unlike Tempa·DOT, the NexTemp thermometer is reusable and automatically resets itself after use.
Accuracy: ±0.2ºF or ±0.1ºC
Shelf Life: 5 years from the date of manufacture
Storage: No climate controlled storage required
Value: No time or resources spent on device maintenance
Range: 96.0°F to 104.8°F or 35.5°C to 40.4°C
Safety: Non-toxic, mercury-free, latex-free
Infection Control: One disposable thermometer per patient
Maintenance: Non-breakable, No calibration or batteries required
NexTemp’s accuracy is the same as, or better than, other thermometers commonly used in a healthcare setting. Each of the dots has its own formulated temperature point that fires at a specific temperature. When a dot fires, the color changes from green to black. The number of temperature dots that fire is dependent on the patient’s temperature. The last black dot provides an accurate temperature reading.
It is important to remember that axillary temperature readings are routinely lower than corresponding oral temperature readings. This difference is normal and expected.
Directions for Use
- Unwrap the thermometer and place it in the mouth under the tongue into either heat pocket for 60 seconds.
- Remove thermometer after 60 seconds
- Read the temperature by looking at the last black dot and discard after use.
Note: The signal will reset 20 seconds after being removed, so the temperature must be read immediately. If the thermometer resets itself before the temperature signal can be read, simply retake the patient’s temperature using the same thermometer. For example, the thermometers displayed read 98.6°F and 38.7°C. After 20 seconds, the thermometer will reset.